Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||28 December 2013|
|PDF File Size:||15.81 Mb|
|ePub File Size:||14.57 Mb|
|Price:||Free* [*Free Regsitration Required]|
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
The forums emoticons took over for a bit there. Message 2 of 6. Message 5 of hp41991a. Sorry, that last line of code there is supposed to read: However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Accepted by topic author Matt C.
Good luck with it Message 6 of 6. I think the would probably take the old commands, but the ALSO has new commands.
4191A RF IMPEDANCE ANL
If not, what would creating a driver for it entail? Hp4191z helps you quickly narrow down your search results by suggesting possible matches as you type. Should I be looking at doing it with LabView or something else? Thanks for your help guys. You really need the manual.
Message 1 of 6.
Most Active Software Boards: It’s 84MB, so I hope you have a fast link. Has anyone done anything on interfacing with the A since Matt C’s last post? IIRC, the sweep parameters were something like this: Guess you never know until you try! Would you mind posting the code you used on the A? When I gave it a try, though, it worked fine. I have not worked with that instrument, but the Agilent web site has the operating and service manual here.
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Message 4 of 6. Matt – you’re right, majual commands for the A are completely different from the – probably because of its age.
In the code above, MHz is an arbitrary frequency. I suspect that this difference might be causing my problems.
Message 3 of 6.